The use of a radiometric calibration target enables the software to calibrate and correct the images' reflectance according to the values given by the calibration target.
The use of a radiometric calibration target enables the software to calibrate and correct the images' reflectance according to the values given by the calibration target. It thus takes into consideration the illumination conditions at the date, time and location of the image capture as well as some of the sensor's characteristics. The calibration target also enables to have an absolute reference, which allows to get absolute reflectance values and make it possible to compare data coming from several cameras or flights. It is recommended to use a calibration target when generating reflectance and index maps.
Pix4Dmapper supports the following calibration targets:
|Calibration target||Automatic importation||Automatic marking||Manual marking|
|Airinov Aircalib||X||X (with Sequoia and MultiSPEC4C cameras)||X|
|MicaSense Calibrated Reflectance Panel (old version)||X||(X) (still need to fill reflectance values)||X|
|MicaSense Calibrated Reflectance Panel (new version)||X||X (with RedEdge, RedEdge-M and Sequoia cameras)||X|
Depending on the type of calibration target and camera used, the workflow to import and mark the calibration images can be different:
- Automatic importation: The pictures of the calibration target are automatically recognized by the software when importing the calibration images along with the other images during the project creation. For more information: Radiometric calibration target.
- Automatic marking: The calibration target region is automatically marked on calibration images and the reflectance values associated to each band are automatically entered. For more information: Radiometric calibration target.
- Manual marking: The calibration target region on calibration images should be manually marked and the reflectance values associated to each band should be manually entered. For more information: Radiometric calibration target.
- The target should be level with the ground, and not at an angle.
- The target should be aligned to the North, to avoid artifacts when sun angle correction is applied.
- The target should not be affected by the shadow or reflection of eventual surrounding objects that would false the image reflectance.
- The calibration region should not be over/under exposed in any of the bands. We recommend placing the target on top of a fairly reflective surface for the auto exposure not to overexpose (e.g. on light asphalt).
- If the images are taken from the ground the camera holder should stand around 1 meter away from the target and not be facing the sun.
- Ensure that the camera GPS writes correctly the time and location information for the processing of the calibration images.
- Take several captures of the calibration to ensure to get at least one valid calibration image. It helps in case of overexposure: Radiometric calibration target.
- Check the calibration target shots while on the field to assess the quality of the images and avoid under/overexposure.
Should the radiometric calibration images be taken before or after flying?
The radiometric calibration image can be taken immediately before or after each flight or during the flight:
- Before or after flying: It does not matter if it is taken before or after flying. It is important that the weather and lighting conditions are as similar as possible to the ones during the flight.
- During the flight: The target should be big enough to be properly visible for the UAV and cover many pixels on the capture.
Should the radiometric calibration images be taken from the ground or while flying?
The radiometric calibration target should cover enough pixels to get good statistics. If a target is big enough to be properly visible from the air, then the image can be taken from the air. If this is not the case, it is recommended to take the image from the ground.
Automatic importation and marking
In possession of both radiometric calibration target and camera supporting automatic importation and marking (for more information: Radiometric calibration target), the pictures of the target are automatically recognized by the software and the target region and appropriate reflectance values are saved in the project. To calibrate the project, simply import the calibration target images along with the other images during the project creation: Step 2. Creating a Project. The reflectance values can still be reviewed in the processing options: Menu Process > Processing Options... > 3. DSM, Orthomosaic and Index > Index Calculator.
Manual importation and marking
To manually calibrate the project images with the radiometric calibration target information, follow the instruction described below:
1. On the menu bar click Process > Processing Options..., the pop-up Processing Options appears.
2. Click 3. DSM, Orthomosaic and Index and select the tab Index Calculator.
3. In the section Radiometric Processing and Calibration, the different available discrete bands are displayed. Under each band, click Calibrate...
Click Browse... to import the corresponding image of the calibration target for the selected band.
Draw a region on the image that will define the radiometric calibration area, the region should be well inside the gray target and exclude all borders.
Insert the Reflectance Factor value for the selected band. The Reflectance Factor value should be furnished by the target manufacturer. Only values between 0 and 1 will be accepted.
Click OK. The camera is now calibrated for this band.
4. Repeat the same procedure for each of the discrete band, click OK and run the project. Note that the radiometric calibration of the images is performed during Step 3. DSM, Orthomosaic and Index.
How to use calibration target on Pix4D Cloud
In order to use calibration target images on Pix4D Cloud the two following workflow are applicable:
- Cloud automatic marking:
In possession of both radiometric calibration target and camera supporting automatic importation and marking (for more information: Radiometric calibration target), import the calibration images along with the other project images. The pictures of the target are automatically recognized by Pix4D Cloud and the target region and appropriate reflectance values are automatically saved in the project.
- Cloud manual marking:
1. Create a project on Pix4D Desktop. For more information: Step 2. Creating a Project.
2. Import the calibration target images along with the other project images during the project creation ensuring that they are all be contained in the same folder.
3. If the target images are not automatically marked, mark them manually. For more information: Radiometric calibration target.
4. Upload the project to Pix4D Cloud. For more information: How to upload project files from desktop to cloud.
What to do if the error message "Unable to compute radiometric target calibration" appears?
The error message:
[Error]: Unable to compute radiometric target calibration, appears because the area of the calibration target is overexposed in the calibration target image. In that case, none of the pixels in the target area can be used for radiometric calibration. To avoid such an issue in the future, see the recommendations on how to take images of the reflectance panel: Radiometric calibration target.
In case of overexposed calibration images, the solutions are the following:
- If another calibration image has been taken during the same flight mission it can be used to replace the overexposed one.
- Redo the flight and acquire new images of the calibration target.
- Process without using the calibration target images (and thus for all discrete bands). Note that without the use of calibration target images, the comparison with other flights will not be accurate.